Yes, you can also take blood samples from the palm, forearm, upper arm, thigh or calf. *Always check with your doctor before performing alternative site tests.
Food, medication, illness, stress and exercise can affect blood glucose levels. Capillary blood at the fingertips reflects these changes faster than capillary blood from other sites. Thus, when testing blood glucose levels during or immediately after a meal, physical exercise, or any other situation, take a blood sample from your finger only.
．Pre-meal or fasting state (more than 2 hours after the last meal).
．Two hours or more after taking insulin.
．Two hours or more after exercise.
The meter stores the 450 most recent blood glucose test results along with respective dates and times in its memory. The meter memory is activated when the meter is switched off.
Push “Main button”.
will appear on the display. Push “Main button” again and the first reading you see is the last blood glucose result along with the date, time and the measurement mode.
Push “Main button” to recall the test results stored in the meter.
Push and hold “Main button”
|2.||Push “Main button” to review 14-, 21-, 28-, 60- and 90- day average results stored in each measuring mode in the order of Gen, AC and then PC.|
Exit the meter memory.
|NOTE:||．Any time you wish to exit the memory, keep pushing “Main button” for 5 seconds or leave it alone for 3 minutes. The meter will switch off automatically.
．Control solution results are NOT included in the day averages.
．When using the meter for the first time, “---“ is displayed when the test results are recalled or the average result reviewed. This indicates that there is no test result in the memory.
The meter displays “On” or “OFF” and . To set the first reminder, push “Main button” to select “On”, then push SET to set the hour. When the hour is flashing, push “Main button” to add the hours. Push SET to confirm and then push “Main button” to add the minutes. Hold “Main button” to accelerate the hours and minutes. Push SET to confirm and go to the next alarm setting.
If you do not want to set an alarm, push SET to skip this step. If you want to turn off an alarm, find the alarm number by pushing SET in the setting mode, push “Main button” to change from “ON’ to “OFF”.
You can use the meter with a USB cable and the Health Care Software System to view test results on your personal computer. To learn more about the Health Care Software System or to obtain a USB cable separately, please contact your local customer service or the place of purchase for assistance.
To transmit data, follow the instructions provided with the software. Results will be transmitted with the date and time. Remove the cable and the meter will automatically switch off.
Replacement lancets and test strips are distributed through major distributors to your nearest stores and pharmacies, Please contact your local customer service or the place of purchase for assistance.